We are treating this matter with the highest possible seriousness, and are working to address this issue as efficiently and thoroughly as possible.
As a result of extensive ongoing analysis, on June 14, 2021, the company issued a recall notification for specific affected Continuous Positive Airway Pressure (CPAP), Bi-Level Positive Airway Pressure (BiLevel PAP) devices, and Mechanical Ventilators.
The recall notification (U.S. only) / field safety notice (International Markets) informs customers and users of potential impacts on patient health and clinical use related to this issue. Possible health risks include exposure to degraded sound abatement foam, for example caused by unapproved cleaning methods such as ozone, and exposure to chemical emissions from the foam material.
Philips is notifying customers and users of affected devices that the company will replace the current sound abatement foam with a new material that is not affected by this issue. Affected devices currently will be either replaced with a new or refurbished unit that incorporates the new material, or repaired to replace the sound abatement foam in customer units. The new material will also replace the current sound abatement foam in future products.
Philips is recommending that customers and patients halt use of ozone-related cleaning products, and adhere to their device Instructions for Use for approved cleaning methods.
Additionally, Philips is reminding customers and patients to review the age of their BiLevel PAP and CPAP devices, as they are typically recommended to be replaced after five years of use.
The company has dedicated significant resources to address this issue, and has developed a comprehensive plan for this correction, and has already begun this process. This effort includes wide-scale, global ramping up of manufacturing, repair, services, supply chain and other functions to support the correction.
Philips deeply regrets the inconveniences caused by this issue, and we are dedicating significant time and resources to give affected patients and customers the service they expect and deserve as we resolve this matter as our top priority.
For more information on the recall notification, as well as instructions for customers, users and physicians, affected parties may contact their local Philips representative or visit www.philips.com/SRC-update. Call 877-907-7508 if you cannot visit the website or do not have internet access.